Part 2 of our series on data logging with the PG10 digital pressure gauge covers data logging based on the sample rate. This allows you to take consistent samples at predefined intervals. Check out the video below:
Narrator: You’re watching APGTV. I’m Shaw Merrill, and today we’re going to show you how to use the data logging feature based on sample rate on the PG10 digital pressure gauge.
To set up continuous logging on the PG10, press the mode button to enter the main menu.
Next, press the up or down arrow button until you see DATLOG displayed on the screen and press enter.
Now press the up or down arrow button until you see LOG displayed and press enter.
Press the up or down arrow button again until you see CONTIN displayed and press enter to enable continuous logging.
Now press the up or down button to cycle through the options until SAMPLE is displayed and press enter.
You can scroll through the sample rates by pressing the up or down button. When you see the sample rate you want, press the enter button to select it.
The PG10 will store a maximum of 60 records stored in a first-in-first-out buffer.
To exit the menu, press the up or down arrow button until you see exit displayed and press enter.
For more on data logging on the PG10 digital pressure gauge, check out our other videos: parts 1, 3, and 4 on our website.
Thanks for watching our quick PG10 data logging video. If you have any questions, please contact us at 888-525-7300, or send us an email to firstname.lastname@example.org. Until next time, I’m Shaw Merrill, APGTV.