In this video, you'll learn how to navigate to and use the tare feature on the PG7 digital pressure gauge. The tare function is used to set the current pressure to zero temporarily. Tare is commonly used to measure a change in pressure during a specific task or event.
Hi and welcome to APGTV, I’m Shaw Merrill and today we are going to show you how to enable the TARE feature on the PG7 digital pressure gauge.
To turn the TARE feature on, press the On/Off button to enter the menu.
Press the zero button to scroll through the options until TARE is on your display, then press the select button.
Now press the zero button to scroll through the options until you see On, then press the select button.
To exit the menu press the zero button until you see exit, then press the select button.
Remember; do not disconnect the gauge from the pressure fitting while the tare function is enabled. The gauge could still be under pressure even though the reading shows 0.
Thanks for watching our quick PG7 menu demo. If you still have questions contact us at 888-525-7300 or send us an email at email@example.com. Until next time, I’m Shaw Merrill, APGTV.