Every now and then, a pressure transmitter just needs to be re-zeroed - especially if it's a sealed unit. But on most submersible pressure transducers, this isn't an option. Not true for the PT-500 from APG. We've given you a way to re-zero your sensor quickly and easily with a simple magnet. Check it out!
Narrator: Our PT-500 submersible pressure transducer is the only one on the market with a field-adjustable zero. This let’s you correct your sensor as it naturally drifts over time. We’re going to show you how this is done.
We have a small tank with 20 inches of water. At this depth, our output should read 5 milliamps. But, you’ll notice our sensor is incorrect.
So we’ll simply:
- Pull our sensor out of the tank
- And use our magnet to adjust the zero back to 4 milliamps.
- If you hold the magnet near the middle of the sensor, it moves the zero up.
- If you hold the magnet near the bottom of the sensor, it moves the zero down.
- Once your sensor is properly adjusted, lower it back in the tank.
All finished! It’s quick, it’s easy, and you’ll only find it on the PT-500 submersible pressure sensor from APG.
Ready to check out our submersible pressure transducers? Do it by clicking on the green button below! See how awesome our submersible pressure transducer really is and how it can help you: