In this video, Shaw Merrill of APG will be showing you how to program your LPU-2428 Ultrasonic Sensor using the submersible option in the LPU-2428 programming software.
The submersible option is used when you are replacing a submersible sensor or pressure sensor such as the PT-500 Submersible Pressure Transducer with a LPU-2428 Ultrasonic Level Sensor.
00:20 To begin programming the LPU-2428:
Wait until the progress bar completes before moving to the next step.
00:39 Now we are ready to setup the remaining parameters.
01:14 Your sensor is now setup and ready to go.
01:17 The remaining advanced features in the programming software will default to automatic settings. To change these settings, please refer to the LPU-2428 user manual programming section.
01:26 Thanks for watching our quick LPU-2428 programming demo.
01:41 Until next time, I'm Shaw Merrill, APGTV.
For more videos on the LPU-2428, visit the LPU-2428 Ultrasonic Level Sensor Playlist.